Answered Your Question? 1 2 3 4 5 Document needs work? You would think that with all of the timers, they could run tasks in parallel, but then the signal routing definitions of those timing signals would become ambiguous. If you need access to this channel, use the generic version of this driver. 400673 QERR_MISSING_URI_OPTION_NAME No option name was found even though a value was specified. 400674 QERR_ENCODER_QUADRATURE_MODE_NOT_SUPPORTED The selected Refer to the documentation for your card. 400383 QERR_CONFLICTING_DIGITAL_DIRECTIONS A digital channel was specified as both an input and an output. this content
If you would like to post the portion of your code that you are having problems with, perhaps there is something that we might see that is causing this problem. Error -50808 : It occurred when you used DAQmx 9.4 or above. Just try to take two slaved DAQ boards and make one task which includes all of the channels on both boards. The numbers you have posted are not DAQmx error codes that I have ever seen before. https://forums.ni.com/t5/Multifunction-DAQ/Is-there-a-document-that-lists-error-codes-for-NIDAQmx/td-p/391117
All rights reserved. | Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishChinese(China)SpanishJapaneseFrenchGerman 29 ratings: 3.58 out of 5   Why do Separate options from the hostname and port using a '?' rather than a '/', much like an HTTP query. 400553 QERR_UNSUPPORTED_BAUD_RATE The given serial baud rate is not supported. 400554 QERR_ETHERCAT_MASTER_ALREADY_RUNNING It must point to an appropriately-sized integer. 400505 QERR_STREAM_FORMAT_NOT_DEDUCED The character format of the stream can only be deduced from the first bytes in the stream. 400506 QERR_MISMATCHED_STREAM_FORMAT The character format I am using a PXI-6225, which is an M Series device.
Members 92 3,838 posts Location:Lafayette, CA Version:LabVIEW 2014 Posted August 15, 2007 QUOTE(jdunham @ Aug 13 2007, 04:42 PM) In my app, I needed long term timed acquisition on some of I ended up using one task, and reading the channels independently. Try using a different clock frequency (usually slower) or a different clock. 400393 QERR_DUTY_CYCLE_NOT_POSITIVE The specified duty cycle is negative and negative duty cycles are not supported by this board. 400394 Labview Visa Error Codes I opened a support request with NI at the same time (2 hours ago) that I started this thread on LAVA.
Solution: Error -223022:This error indicates that the ADCs never came out of their initial calibration cycle which is done immediately after driver load when the system boots. Ni Error Code This configuration will work for either sample mode and if used in conjunction with the DAQ Task State Model will improve driver performance. One common occurrence of this error is when a DAQmx Read function is used inside of a while loop when the task is configured for finite acquisition. http://digital.ni.com/public.nsf/allkb/4159A4241B8A24B3862570EC007AB6B7 The card or driver may not be installed.
I was able to find an NI knowledge base entry, Causes of NI-DAQmx Error 50103 "The Specified Resource is Reserved", which addresses this issue. Labview Error 1073807339 To avoid leaving no means to connect to the target, the last URI may not be removed. Try rebooting, requesting fewer samples, or adding more memory to your machine. 400355 QERR_DRIVER_INCOMPATIBLE_WITH_BOARD_DLL The board-specific HIL driver passed an invalid parameter to the operating system specific kernel-level driver for the Channel numbers are typically divided into ranges according to functionality.
This design requires that the Daq reader object, knows the fastest sampling time, so each object (when they were created) told the Daq reader object it's fastest sampling time. //Mikael Share check my blog My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. Ni Daqmx Error I ended up using one task, and reading the channels independently. Labview Visa Error -1073807339 There are two solutions to this problem.
In this case, contact the driver manufacturer. 400726 QERR_HIL_TASK_FLUSH_NOT_SUPPORTED The hil_task_flush function is not supported by this particular card. news Valid values for the "peer" option are "one", "any" and "broadcast". 400524 QERR_CANNOT_BROADCAST The communication protocol, such as UDP, could not be configured for broadcasting. 400525 QERR_ETHERCAT_VALIDATE_DATA_WRONG_LENGTH The Validate/Data field in Because the DAQmx task is stopped and restarted during each loop iteration, placing the task back into the committed state for the next loop will ensure it is not having to The wiimote is possibly already opened by another process. 400650 QERR_DAQMX_CANNOT_CLEAR_TASK DAQmx was unable to clear or release a National Instruments task. Labview Error 200279
I opened a support request with NI at the same time (2 hours ago) that I started this thread on LAVA. You can turn this off, but the default is on. It makes troubleshooting my code very difficult as I the only way I can find out what is going on is to post here where I might have to wait a have a peek at these guys Architecturally it was annoying because I had to know about all possible tasks in one VI and configure them all before doing any of the reads.
You could even configure the AI task to start together with the AO task by making it a slave of the AO clock. Labview Error 50103 Multifunction DAQ Register for the community · Log in · Help ForumsCategoryBoardDocumentsUsers turn on suggestions Auto-suggest helps you quickly narrow down your Telegram is corrupted! 400580 QERR_INVALID_ETHERCAT_MASTER_STATE EtherCAT master cannot be set to the state requested.
NitishLo Member 01-12-2016 03:41 PM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Link below gives you driver code Found under Help/Explain Error and enter that code number in. -AK2DM ~~~~~~~~~~~~~~~~~~~~~~~~~~"It’s the questions that drive us.”~~~~~~~~~~~~~~~~~~~~~~~~~~ 0 Kudos Message 5 of 7 (1,487 Views) Reply 0 Kudos Re: Is Members 92 3,838 posts Location:Lafayette, CA Version:LabVIEW 2014 Posted August 14, 2007 Hello Everyone, Thanks for the responses! Labview Custom Error Codes Try again later. 400415 QERR_HOST_SHUTDOWN The host was shut down during the operation. 400416 QERR_CONNECTION_RESET The connection was reset.
Poor|Excellent Yes No Document Quality? This is a simple description for the above code. 1) DAQ Assistant can make a DO signal. So far I've only gotten automated responses from NI stating that someone will respond to me within 6 hours. check my blog If you find a different way, it would be great to hear about it, but I gave up.
Is there a document that lists error codes for NIDAQmx? Try restarting LabVIEW. 400651 QERR_DAQMX_CANNOT_CREATE_TASK DAQmx was unable to create a new task. Generated Thu, 20 Oct 2016 01:56:24 GMT by s_nt6 (squid/3.5.20) If no devices are open, try restarting LabVIEW. 400654 QERR_DAQMX_ERROR_SAMPLING_ADC DAQmx failed to execute the National Instruments analog input task. 400655 QERR_DAQMX_ERROR_RESETTING_DEVICE DAQmx failed to reset the device. 400656 QERR_DAQMX_CANNOT_ATTACH_DAC_TO_TASK DAQmx
To access the channels independently (after the global task start), just set DAQmxRead.ChannelsToRead immediately before calling DAQmx read. Related Links: KnowledgeBase 4BHID6DE: Why Are NI Device Loader (nidevldu) and NI Configuration Manager Services Not Starting and Giving an Error -88705KnowledgeBase 7BDCQQNN: Error -89130 When Using Test Panels or DAQ DAQmxGetErrorString will return a message that corresponds to the DAQmx error number, but the numbers you have are not DAQmx error codes. IMO, this is either a bug or a very poor design of the DAQmx API/behavior.
DAQmxGetErrorString doesnt work. Try using one of the more specific hil_read functions, HIL Read blocks or HIL Read polymorphic instances. 400680 QERR_HIL_READ_ANALOG_NOT_SUPPORTED The Analog polymorphic instances of the HIL Read VI are not supported My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. Does anyone else have experience with or thoughts about this issue?
Either way, I'll post updates here if I learn anything from NI. When I did something like this, I did try to use separate tasks and never got it to work. Many error code text files are found in [National Instruments]\Shared\Errors as shown below.The error codes for the Modular Instrument drivers, which include drivers such as NI-SCOPE and NI-DMM amongst others, can Due to architectural considerations, we have decided to use separate DAQmx tasks for reading each channel.
Share this post Link to post Share on other sites LAVA 1.0 Content 1
Error Codes in Numerical Order Error Code Symbolic Name Description 400241 QERR_OUT_OF_MEMORY Not enough memory to perform the operation. 400242 QERR_OUT_OF_RESOURCES Not enough system resources are available to perform the operation. My solution is to create a wrapper VI around DAQmx Read which implements a retry (in a While Loop) when error -50103 occurs, until the timeout expires. So, I need to look for nidaqmx.h in my computer (if it was installed) to get only the reference name of the error code??Why it is not in the help files??Very Answered Your Question? 1 2 3 4 5 Document needs work?