An exact target faradaic efficiency can be obtained optimizing η, which in general requires a detailed electrochemical model to calculate ε as a function of current and methanol concentration; for simplicity, Figure 1 The process layout of the reference system . I have one more problem. DMA is not allowed. http://softacoustik.com/labview-error/labview-error-6.php
I have one more problem. Stack temperature is measured with a K-type thermocouple, 50 mm deep inside of the middle bipolar plate. Also, the driver attempts to achieve complicated gain distributions over SCXI channels on the same module by manipulating the scan list and returns this error if it fails. -10400 userOwnedRsrcError The The inbuilt DIO Config VI has a group direction input , i dont know how but the dio config that i use doesnot have group direction input.... http://forums.ni.com/t5/LabVIEW/NuDAQ-acquisition-using-DAQ-2204-Error-10409/td-p/1706584
whenever i change some value on the vi and try to rerun it an error comes at digital group congif (error 10409) with reason -The specified group is in use. Your cache administrator is webmaster. Try different settings or use AutoAssign in the NI-DAQ Configuration Utility. -10255 remoteChassisDriverInitError There was an error in initializing the driver for remote SCXI. -10256 comPortOpenError There was an error in
energies-08-10409-t001_Table 1 Table 1 Model equations and parameters of the reference model . The concentration overshoot at the beginning of the simulation does not appear in the experimental results of Figure 9 because of the presence of liquid in all system elements and the Two equations must be solved simultaneously to determine V ˙ fuel and V ˙ sol for λ an and inlet concentration control. The main features of the model will be recapitulated in this section. 3.1.1.
Loading... Try using a slower waveform update rate if possible. -10884 pretrigReorderError Could not rearrange data after a pretrigger acquisition completed. -10920 gpctrDataLossError One or more data points may have been lost Sixclear 3,375 views 9:23 VI High 44 - Learn How to Write to a Text File with LabVIEW - Duration: 5:47. of scan value in between 10 to 1000 then error ...
Consult the device user manual to determine group ramifications with respect to DMA. -10458 diskFullError The storage disk you specified is full. -10459 DLLInterfaceError The DLL could not be called due This excessive fuel supply results in a higher concentration in the experiment than that in the simulation, as shown in Figure 9c,d. Loading... In this condition, the inlet concentration c ¯ in an is calculated to be smaller than c ˜ mix because the inlet concentration is proportional to the current, per Equation (13).
Sixclear 1,517 views 8:02 VI High 11 - How to Use "Suspend When Called" in SubVI Node Setup in LabVIEW - Duration: 2:38. United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Hardware Boards : Digital I/O : Clearing error 10409 using Traditional DAQ and DAQCard 6533 Digital This allows one to blend the solution from the solution tank with water, in order to produce an anode inlet concentration lower than the one in the solution tank: this can Close Yeah, keep it Undo Close This video is unavailable.
The three control objectives are λ an control, controlling c in an and maintaining the level of the solution tank. http://softacoustik.com/labview-error/labview-error-ni-488.php Have you got any suggestion on how to set a triggered acquisition from 4 different DAQcards, using as trigger the 1st channel on card1 for all of the 4 cards?? Watch QueueQueueWatch QueueQueue Remove allDisconnect Loading... Babidi Member 09-15-2011 08:54 AM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Hi everybody, I am trying
Working... The rising solution weights at the beginning in Figure 10a,b are attributed to the initially low stack temperature, which allows more water to condense. If you are allowing the driver to calculate the analog input channel clock rate, the driver bases the clock rate on the board type; so you should check that your board this content the product is not an NI device or the drivers does not support.........
Loading... Sign in to add this video to a playlist. In addition, the fuel and water flows will need to be dosed, so that the concentration of the resulting solution is c ¯ in an , so that the solution volume
Active Participant 04-03-2009 09:43 AM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Hi Ryan,This error is likely coming The electrochemical efficiency ε has been previously studied , establishing that there is an optimal concentration at which voltage is maximum: higher concentrations promote more crossover (increasing the cathodic overvoltage), and This is extremely frustrating, as my laptop is slow and it takes about 3 minutes to shut it down and restart it. Up next VI High 47 - Learn How to Create Error Handlers and Implement Error Handling in LabVIEW - Duration: 7:26.
The latest version supported in LabVIEW 6.1 is NI-DAQ 6.9.3 available at:http://digital.ni.com/softlib.nsf/websearch/50F49E43A22C0D2186256BF3006B9A96?opendocument&node=13207...Let me know if none of these approaches fixes the problem.Regards,Neil S.Applications EngineerNational Instruments 03-12-2007 10:41 AM Give Kudos 0 Nestor Ceron 34,021 views 7:10 VI High 56 - How to Pass Data Between States in a LabVIEW State Machine - pt 1 - Duration: 6:44. hi.. have a peek at these guys what can be the cause of this problem .
Faradaic Efficiencies As shown in Figure 11a, the efficiency of the two-mixer system is marginally higher than that of the separate tank system, while both are significantly higher than the reference For example, the buffer is at an odd address, is not aligned to a 32-bit boundary, is not aligned to a 512-bit boundary, and so on. i m using an external clock signal that comes from I2S bit clock signal. the reason was that the while loop termination was not correct.i just changed from stop if true to contiue if true.
Try using a slower data acquisition rate if possible. -10883 wfmPollDataLossError New data could not be transferred to the waveform buffer of the remote SCXI unit to keep up with the Sign in 1 Loading... Water and fuel flows are also used to control the solution tank’s level, and those components are indicated with the subscript V . You can switch between the two libraries only by running the DAQ VI Device Reset before calling _DAQ compatibility VIs or by running the compatibility VI Board Reset before calling DAQ
Please try the request again. I am not sure why you have two separate DIO Config Vis. the inbuilt vi dont show any error if i remove the control that is meant to be connected to the group direction input of the dio config vi.