The problem seems to be related to the nidevldu since it refuses to be stopped or restarted. I can't believe it, Thank you!!!” Delcie- Yesterday “I spent all day trying to sort this out then found your site. Keep me posted. 0 Kudos Message 10 of 34 (2,562 Views) Reply 0 Kudos « Previous 1 2 3 4 Next » All Forum Topics Previous Topic Next Topic Privacy | One thing was remarkable: The uninstallation failed on the first attempt, it could not stop the services. http://softacoustik.com/labview-error/labview-error-6.php
Subject: Re: DAQmx error -88705 From: "Geoffrey Schmit"
A >> fatal status of -88705 indicates that the driver for the specified >> device is not present. >> > Yes. All rights reserved. Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishJapaneseSpanishFrenchPortugueseKorean 62 ratings: 3.95 out of 5   Why Are NI Any other ideas? 0 Kudos Message 7 of 34 (2,578 Views) Reply 0 Kudos Re: Error 88705 with a USB-6366 DAQ Carmen_Martinez Member 02-07-2012 03:31 AM Options Mark as New Bookmark STEP 2:Click the "Quick Scan" button.
I hope this helps. To start it properly go to Start >> Control Panel >> Administrative Tools >> Services and select the NI Device Loader Service. The VI version uses the > same NI-DAQmx VIs. > > I can also deploy the exe to that PXI from host, and works fine. The prosoften make house computer repairst boot or not service privacy and thing systently the facturn you choose the computer repair and MAC OS, which is assessions malfunction.
How can I resolve this issue? KnowledgeBase 491FPHB9: USB Device Firmware Incompatibility with DAQmx Base 3.0 or Later KnowledgeBase 5XG8M1RP: Why Does My DAQ or Simulated DAQ Device Fail the MAX Self-Test With Error -88705? I'm unable to use most of my PXI cards, 5922, 6052E, 6115, 4071, etc. The device does pass selftest on another machine, it clearly is a software issue and several things point to the problem being the Device Loader service (it is in the state
NOTE: If you have MAX version 4.2 the location of Data folder has changedFor Windows XP and 2000:C:\Documents and Settings\All Users\Application Data\National Instruments\MAX\DataFor Windows Vista:C:\ProgramData\National Instruments\MAX\Data Restart the computer. Answered Your Question? 1 2 3 4 5 Document needs work? United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Hardware Boards : Multifunction DAQ : Error 88705 with a USB-6366 DAQ Multifunction DAQ Register for Additionally, you may need to run lsdaq and/or close and re-open LabVIEW to reset the software.
To properly starttheservice and ensure that the service is always started at startup follow these directions: Go toStart » Control Panel» Administrative Tools» Services Right click onNI Device Loader Service(may be http://digital.ni.com/public.nsf/allkb/B179266E288CE329862573B40078DDC9 Please try restarting your computer. 3) Firmware Update NeededDuring a driver installation, firmware updates download with the driver for many NI DAQ devices. I spent hours looking for a solution to this error and finally I found one. Cheers!
Poor|Excellent Yes No Document Quality? http://softacoustik.com/labview-error/labview-error-ni-488.php KnowledgeBase 45HGAIJQ: Why am I getting DAQmx Error Code -89130: Device is not Available for Routing?KnowledgeBase 5NO8O2DO: Repairing NI Software Installs Leaves NI System Web Server Unregistered Drivers and Updates:NI-DAQmx Attachments: Is it a problem in the new MAX version?? Thanks dearly, Michael Abhinav T 2007-06-12 21:40:19 UTC PermalinkRaw Message Hi Michael, This error message is returned because the "NI Device Loader" Windows service is Restarting LabVIEW or unplugging and re-plugging the device back into the USB port may fix this problem.Error -6006This NI-DAQmx Base error can occur when an incompatible version of NI-DAQmx is installed
geoff -- Geoffrey Schmit Senior Software Engineer National Instruments [email protected] www.ni.com For more informaton and to subscribe to the email list and a topic linked search of archives at "http://sthmac.magnet.fsu.edu/infolabview" I'm not a PC guru by any stretch, but this was a godsend. The device may not be installed on this system, may have been unplugged, or may not be installed correctly. have a peek at these guys Did you modifiy something in your computer?
Right click on it select "properties" and then set the startup type as "Automatic" and click the start button. Worked a treat!” Cherelle- 1 Month Ago “Thanks for sharing, I no longer have to put up with the dreaded Labview Error 88705” Sommer- 2 Months Ago “Will this work with Attachments: Report Date: 05/17/2012 Last Updated: 01/23/2013 Document ID: 5XG8M1RP Your Feedback!
Get Your PC Running Normal Again in Under 10 Minutes Filed in Windows Windows Labview Error 88705 on Problem with Windows Labview Error 88705? It's important to scan your PC every now and again to ensure that these files are in place and everything is as it should be. For example, DAQmx Base 3.1 for Windows is only compatible with NI-DAQmx 8.6 for Windows. If it is not, use theStartbutton to start the service.
STEP 2: Click "Quick Scan" Button to Scan Your Computer. I also set the Recovery options of the NI Device loader Sevice to "Restart", still no luck. Anybody know what's going on? check my blog Then rename Data folder from C:\Program Files\National Instruments\MAX\ directory.
Select Start»Control Panel»System and Security»AdministrativeTools»Services. I acquired a NI US-6211 about a month ago, installed with DAQ-MX 8.5. Have you ever succeed using your USB-6366 DAQ or you were never able to pass the self-test? I cannot tell after what software installation it stopped working, and that is exactly my main issue, i have no idea what interfers.
Yes, after installing the DAQ software the first time i tested it and the selftest passed. Since first install, twice my device stopped being recognised (also faild selt test with -88705), with errpr event in Windows event log: " "The NI Device Loader service terminated unexpectedly. http://digital.ni.com/public.nsf/allkb/86256F0E001DA9FF86256FFD005B827C?OpenDocument Hope this finally solve your problem. Visitor Comments 8 Comments for "Want to Repair Labview Error 88705?" Wesley - Today “This Repaired the Labview Error 88705 message.
Unplugging and re-plugging the device back into the USB port may fix this error in those situations. In MAX, NI-DAQmx Devices, performing the Self Test yields an error message "Error -88705 occurred at Self Test. ...The specified device is not present or not active in the system." However, Can you install MAX and DAQ-mx in other computer to see if your device can pass the self-test in other machine? Error -6003This error may arise when an incompatible version of NI-DAQmx is installed on the same system.
My PC is now running much faster and is far more reliable.