You can also select a location from the following list: Americas Canada (English) United States (English) Europe Belgium (English) Denmark (English) Deutschland (Deutsch) España (Español) Finland (English) France (Français) Ireland (English) Please Contact NI for all product and support inquiries. Does your device pass a self-test in Measurement and Automation Explorer and, if so, what happens when you try running a test panel?Regards, 0 Kudos Message 2 of 2 (604 Views) Top contributing authors: Name Posts Jim12345678 2 user's latest post: error 89130 with counters Published (2016-05-05 13:13:00) Tried restarting the PC (strictly a wall powered laptop) etc.. No luck. Took the http://softacoustik.com/labview-error/labview-error-6.php
Based on your location, we recommend that you select: . Answered Your Question? 1 2 3 4 5 Document needs work? Discover... Poor|Excellent Yes No Document Quality? http://digital.ni.com/public.nsf/allkb/61371556441E5FF68625730600775934
Communication with the device was lost. This dependency is removed in DAQmx 9.6 and later. My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
Join the conversation Posts Forums Imdb Yuku Lefora Spring Login | Sign Up Loading... If NI Device Loader still does not respond, check if National Instruments mDNS Responder Service (listed as NI mDNS Responder Service in DAQmx 9.6 or later) is started in Services. Start»Settings»Control Panel»Administrative Tools»Services. https://forums.ni.com/t5/Multifunction-DAQ/DAQmx-Error-Code-89130-Device-is-not-Available-for-Routing/td-p/1009434 Also, what version of DAQmx do you have?
Solution: When a computer's operating system (OS) takes a long time to boot, the additional time used by the OS to address the heavy demand of resources from other services may If it is not, use theStartbutton to start the service. To reset all devices in Traditional NI-DAQ, right-click the Traditional NI-DAQ Devices folder in MAX and select Reset Driver for Traditional NI-DAQ. All rights reserved. Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishJapaneseSpanishFrenchPortugueseKorean 62 ratings: 3.95 out of 5   Why Are NI
Task Name: MWDAT0018 Status Code: -89130 0 Comments Show all comments Tags nidaqdatdaq Products Data Acquisition Toolbox Related Content 1 Answer MathWorks Support Team (view profile) 13,642 questions 13,642 answers 13,641 Right-click on theNI Device Loaderservice (also seen as nidevldu), and selectPropertiesfrom the drop-down menu. More site info... You can do this by following these steps: Windows 8/7 Log in as an administrator or as a user with administrative privileges.
Make sure the service status isStarted. http://softacoustik.com/labview-error/labview-error-ni-488.php Neeraj. 0 Kudos Message 1 of 3 (1,210 Views) Reply 0 Kudos Re: Error -89130 occurred at DAQ Assistant P.Sanders Member 12-15-2010 07:56 PM Options Mark as New Bookmark Subscribe Subscribe If not, I would say it is bad, contact NI for repairs. For SCXI devices, you must reset the communicator DAQ device.
Showing results for Search instead for Did you mean: Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the Top Bookmark Reload the page to see its updated state. Possible Reason(s): Specified channel is not in the task. check over here NIDAQmx error : Device not available for routing.
Learn more MATLAB and Simulink resources for Arduino, LEGO, and Raspberry Pi Learn more Discover what MATLAB® can do for your career. All rights reserved. Changing this configuration should ensure that the correct services start every time you reboot your computer.
Aftera successful reinstall,conduct aself-test and run some Test Panels in MAX. This thread profile page shows the thread statistics for: Total Authors, Total Thread Posts, and Thread Activity Home| About Us| Submit Your Site| Update Your Site| Get Search For Your Site| They will also need daqmx version (may ask to upgrade), ... It is possible that the device needs to be reset or that the device is being reset.
For example, if you have used the device through Traditional NI-DAQ, you must reset the device before the requested route can be made. Please Contact NI for all product and support inquiries. When I look in Windows Device Manager, my device shows up correctly. this content Apache error AH01797 with trow to white screen server error 500 APPCRASH error codes with some programs.
Please tell us why. If you are resetting the device, wait for the reset to complete. Doing so will make it available under NI-DAQmx, as mentioned in the error message.For more information, please see the following related Bug Report which discusses the same source of error:http://www.mathworks.com/support/bugreports/details.html?rp=284483 0 All rights reserved. | Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishFrenchJapaneseSpanish 5 rating: 1.60 out of 5   Why Is
Learn MATLAB today! Thanks. Opportunities for recent engineering grads. APPCRASH error codes with some programs.