This option helps you avoid potential problems the next time you run the VI. The card simply didn't work. Regards,Matt 1 Kudo Message 2 of 14 (299 Views) Reply 1 Kudo Re: Error -1073807343 occured at Visa open RavensFan Knight of NI 09-21-2016 01:42 PM Options Mark as New Bookmark Poor|Excellent Yes No Document Quality? weblink
Check double for baudrate and stop characters and handshake in the manual. tcap 3 user's latest post: Old visa open, error code... Delete the string and make the actual VISA Resource Name visible on the front panel. It allows one to run GPIB commands to an instruments over your network by converting the GPIB string to a network packet and sending it to the ENET100. dig this
GPIB-USB-HS not mounting in CentOS 7 (NI-VISA-15.5.0 installed) USB device shown in NI-MAX, but SOMETIMES VISA Resource... I'd say the main problem here is that I know very little about LabView and VISA. Sign In Sign In Remember me Not recommended on shared computers Sign in anonymously Sign In Forgot your password?
LabVIEW 6.0.1 has a lot of errors in it that were introduced by mistake. I want to uninstall NI-VISA?? To address your instrument, you can either use its VISA Alias or VISA Resource Name. USB?
Using Matt's suggestion of tracing the program did help as the error is thrown when the loop calls a particular com port and crashes at the VISA property's Interface Description. I want "Enable Keysight GPIB Cards for 488 Programs" enabled within Keysight Connection Manager. Published (2015-03-13 08:51:00) Sorry, posting from my phone and Swype replaced last with lady. https://forums.ni.com/t5/Instrument-Control-GPIB-Serial/Old-visa-open-error-code-1073807343-using-VISA-with-a-GPIB/td-p/3103032 Dennis_Knutson 3 user's latest post: Old visa open, error code...
Do you have an NI GPIB controller? Trouble communicating with a GPIB instrument using VISA... NI-VISA should be. 0 Kudos Message 2 of 14 (2,085 Views) Reply 0 Kudos Re: Old visa open, error code 1073807343, using VISA with a GPIB device, and VISA resource names If I just enter the VISA resource name from MAX manually, GPIB0::17::INSTR, and save and try running that, I get the error(code -1073807346): VISA Open in SR715/720 Initialize2.vi->SR715/720 Getting Started.vi
Please try the request again. Almost seems like a chicken and egg scenario as I cannot test if the string is empty because in order to retreive the string, it throws the error. Thank you for the help! 0 Kudos Message 9 of 14 (2,024 Views) Reply 0 Kudos Re: Old visa open, error code 1073807343, using VISA with a GPIB device, and VISA NI-VISA should be.
Pruebe lo siguiente: Haga clic en el botón Actualizar o inténtelo de nuevo más tarde. have a peek at these guys Contact Us Site owned and operated by VI Shots LLC Community Software by Invision Power Services, Inc. × Existing user? Which VISA did you install as primary? Share this post Link to post Share on other sites george seifert 9 Extremely Active Members 9 399 posts Posted May 3, 2007 I just talked to an NI engineer
Here is a picture of what the front panel and block diagram of the sub-VI Initialize2.vi (I changed the name in case I screwed things up, but I'm pretty sure the Sorry, I should have mentioned that explicitly. The "hanging around" tells me that something is not releasing a resource so Windows keeps it around. check over here Loading...
Tomi Unfortunately it doesn't hang on a PC without the hardware (EasyDAQ relay driver). Opening multiple visa sessions 8. So I recommend you to contact the vendor of easyDAQ.
I've seen that error when a non-NI controller is used. He did suspect that it's a LabVIEW problem and that the program is hanging on for dear life and won't let itself be shutdown. Later that day I got a message from NI that the issue was known and could be resolved by downloading and installing a new BIOS for the HP computer according to Do not use the 'Old' VISA Open.
If I had to guess, maybe one thing could be that in MAX, under Tools->NI-VISA->VISA Options->My System->Conflict manager, I have both Agilent VISA and NI VISA enabled; I know sometimes problems Solution: This error can happen for a number of reasons: Incorrect Addressing: Verify that you are addressing your instrument correctly. Is there a particular one that just not look right? this content The issue turned out to be that on *this* computer the controller cards had been reassigned to VXI1.
My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. I have to use the task manager to shut it down. As a result, when you try to access that VISA session using a global variable, you can get an invalid VISA session error if the option is enabled.In LabVIEW 6 and I'm going to try the exe on my development PC tomorrow with the hardware.
This is not the only program which has this problem and I have no Idea what the problem could be. I would make sure that both I/O Connection Library and NI-VISA are the latest versions. Trying to connect PSM1735 NumetrtiQ via RS232 but getting "Error -1073807253 occurred at VISA Read in Serial_connection.vi"Possible reason(s): VISA: (Hex 0xBFFF006B) A framing error occurred during transfer. The error is thrown at the VISA open.
Total authors: 3 authors Total thread posts: 14 posts Thread activity: no new posts during last week Domain info for: ni.com Other posts in this thread: Reply 1: Re: Old visa So (regarding the 1st link) it seems like I had enabled Tulip a while ago, and the thing I just tried was the second thing I needed to enable. I'd I'm referring to the last step of tip 1 here: http://cp.literature.agilent.com/litweb/pdf/5990-3731EN.pdf?cmpid=1273 However, when I try it in Keysight Connection Expert, it's not letting me select that... 0 Kudos Message 5 Third, if you have an NI controller and with the instrument connected, you right click on the controller listing and select Scan for Instruments.